Particulate monitoring requires the measurement of the mass concentration of particulate matter in stack emissions to atmosphere. There is no known method of continuously measuring particulate concentration by direct measurement and therefore, all continuous particulate measurement methods are inferential.
Each method involves the measurement of a parameter related to particulate concentration and then requires a calibration to be determined to convert the measured value into one of particulate mass concentration.
In order to output particulate concentration in mg/m3, a Density Scale Factor (DSF) must be determined. The Density Scale Factor for any given installation will vary based on the type, shape, size and colour of the particles in the gas stream and therefore, the Density Scale Factor must be determined on site, through comparison of the instrument readings with an independent reference measurement*.
During normal operation:
Particulate concentration (mg/m3) = measured parameter x DSF
Dynoptic currently offers three different particulate monitors which each use the open path light transmission technique.
Don’t just take our word for it, read what some of our clients are saying about our products.
Below are our core product applications, simply select them for more information about suitable monitors